Medical Oxygen Concentrator
Face Mask &Aroma Beads
Other medical devices
Protective clothing Instruction for Disposable Medical Protective Coverall Surgical Gown
Protective clothing Instruction for Disposable
Medical Protective Coverall Surgical Gown
[Product Name] Disposable Medical Protective Coverall
[Scope of application] Provide barrier protection for clinic medical staff during contact with
blood, body fluids and secretions coming from patients during operations.
[Main Structure Composition] Made by non-woven material and composed by hood, blouses
[Quantity] 1 piece/bag, 50pcs/carton, Gross Weight: 15kgs, Net Weight: 14kgs
[Size] S L XL XXL 1 6 0 ( S ) , 1 6 5 ( M ) , 1 7 0 ( L ) ,1 7 5 ( X L ) ,1 8 0 ( X X L )
【Product Performance】 1 Appearance
1.1 The protective coverall shall be dry, clean and mildew free, the surface of coverall shall not
allow products self- adhesion, cracks, holes and other defects.
1.2The connect parts of the protective coverall can be processed by stitching, gluing or welding
etc. When stitching, pinhole should be sealed well, and there should be 7-14 stitches per 3cm
length. Stitching should be straight and even without open seam. Gluing and welding part should
be tight sealed without bubble or penetration.
1.3The zipper of the protective coverall cannot be exposed to outside and the zipper shall be
1.4 The structure of the protective coverall shall be reasonable, easy to wear and take off. The
joint part shall be tight and well sealed.
1.5 Elastic closure shall be used for sleeve and ankle opening. Hood and waist part can use elastic,
rope or Velcro closure.
2 Barrier Performance:
2.1 Hydrostatic Resistance
2.2 Moisture Transmission
2.3 Anti-synthetic blood penetration
2.4 Anti-static Performance
2.5 Microbiological Performance
Scan QR code